Prescription is required to purchase this product.
Email a copy to firstname.lastname@example.org or fax to 888-251-1979
Luna G3 BPAP 25A by 3B Medical
Luna G3 BPAP 25A offers a modern streamlined design with a small footprint. The well-lit LCD screen composes a user-friendly interface and conveniences that should be standard on all CPAP, BPAP, BiPAP, and BiLEVEL devices. Clock, Temperature and Humidity Monitor on stand-by for convenience.
- The G3 series has a pre-heat feature to warm your humidifier water up for ultimate comfort
- Intelligent Auto Humidity control with integrated heated tubing allows the Luna G3 maximum moisture while ensuring rain out in the tubing is not an issue.
- New Auto Ramp feature sense when the user falls asleep and will adjust the ramp time appropriately
- Data Transfer via SD Card
- Accessory and supply reminders built into the menu to keep you on top of your replacement supplies
- Universal power supply meaning you can take your Luna G3 anywhere in the world with you.
- Multi-stage noise reduction technology is used to ensure sound performance is under 26db
- Luna G3 QR allows you to track your usage information on your mobile device using a QR code displayed on the screen of the Luna G3. The 3B Luna QR app is available in the Apple or Google Play store for free. Download, Register, and snap a photo of the QR code on the Luna G3 to see your data.
Included in Box
- Luna G3 BPAP 25A
- Power Adapter
- Power Cord
- 2 Washable Air Filters
- User Manual
- Carry Case
- SD Card
- Heated Tubing
- Dimensions (L x W x H): 10.4in x 5.7in x 4.5in
- Weight 3.7 lbs
- Water Capacity: To maximum fill line 360 mL
- Humidifier Settings: off, Auto, 1 to 5 (95°F to 154.4°F)
- Humidifier Output: No less than 15mg H2O/L
Mode of Operation
- CPAP, AutoCPAP, S, AutoS
- The SD card can record patient data and fault information
AC Power Consumption
- 100V - 240V ~, 50Hz / 60 Hz, 2A
Device offer to Heated Tubing Communications Port
- 24V 18W
- 4 to 25 cmH2O (in 0.5 cmH20 increments)
- Under single fault conditions, ≤ 30 cmH20 for CPAP and AutoCPAP mode, ≤ 40 cm for the rest of the modes
- The ramp time ranges from 0 to 60 minutes
Sound Pressure Level
- < 26 db, when the device is working at the pressure of 10 cmH20
The Form and the Dimensions of the Patient Connection Port
- The 22mm conical air outlet complies with ISO 5356-1
- Foam Washable Filter
- 2 years
Regular cleaning of the Luna G3 BPAP 25A and its accessories is especially important for the prevention of respiratory infections. CPAP, BPAP, BiPAP, BiLEVEL sanitizing machines are also recommended.
Cleaning the Water Chamber - Remove the water chamber from the BPAP. Open the top of the water chamber and hand wash with a mild liquid soap, rinse thoroughly and wipe dry. Alternatively a sanitizing machine can be used.
Cleaning the Transfer Box - The transfer box is located between the water chamber and the pap device. Please reference the user manual. It is recommended for the transfer box to be cleaned weekly by hand washing with a mild liquid soap, rinse thoroughly, and wipe dry.
Cleaning the Enclosure (exterior of PAP device) - Wipe the exterior surface with a soft cloth to remove dust as needed.
Instructions depend on which type of filter you are using. Luna G3 is offered with a Disposable Filter or a Reusable Filter. The filters are not to be used together at the same time.
If using Disposable Filter (White) - It is recommended to replace the Disposable filter monthly or sooner as needed if showing signs of wear.
If using Reusable Filter (Black foam)- It is recommended to handwash the filter weekly. Note it is important the filter be completely dry before placing back into the PAP device. The reusable filter should be replaced every 6 months or sooner if the filter is showing signs of wear.
Recommended Replacement Intervals:
Nasal Cushion: 1-3 Months
Mask Frame: 6 Months
Headgear: 3-6 Months
Tubing: 3-6 Months
Water Chamber: 3-6 Months
Reusable Filter: 3-6 Months
Disposable Filter: Monthly
Returns accepted within 30 Day of receiving. Return of CPAP / BiLEVEL devices will only be accepted if in "New" and "Unused" condition. Customers are responsible for any return shipping cost.
Copy of Prescription Required before shipping